ITM Web Conf.
Volume 12, 2017The 4th Annual International Conference on Information Technology and Applications (ITA 2017)
|Number of page(s)||5|
|Section||Session 3: Computer|
|Published online||05 September 2017|
High Precision Infrared Temperature Measurement System Based on Distance Compensation
1 Suzhou Institute of Biomedical Engineering, Chinese Academy of Sciences, 215163 Suzhou, China
2 School of communication and information engineering, Shanghai University, 200444 Shanghai, China
a Corresponding author: firstname.lastname@example.org
To meet the need of real-time remote monitoring of human body surface temperature for optical rehabilitation therapy, a non-contact high-precision real-time temperature measurement method based on distance compensation was proposed, and the system design was carried out. The microcontroller controls the infrared temperature measurement module and the laser range module to collect temperature and distance data. The compensation formula of temperature with distance wass fitted according to the least square method. Testing had been performed on different individuals to verify the accuracy of the system. The results indicate that the designed non-contact infrared temperature measurement system has a residual error of less than 0.2°C and the response time isless than 0.1s in the range of 0 to 60cm. This provides a reference for developing long-distance temperature measurement equipment in optical rehabilitation therapy.
© The Authors, published by EDP Sciences, 2017
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.