ITM Web Conf.
Volume 30, 201929th International Crimean Conference “Microwave & Telecommunication Technology” (CriMiCo’2019)
|Number of page(s)||5|
|Section||Information Technology in Telecommunications (3a)|
|Published online||27 November 2019|
System for receiving and processing diversity acoustic noise control data
1 Murom Plant of Radio Measuring Instruments, 2, Karacharovskoye shosse str., Murom, 602265, Russia
2 Vladimir State University named after Alexader Grigoryevich and Nickolay Grigoryevich Stoletovs, Murom Institute, 23, Orlovskaya str., Murom, 602264, Russia
3 Vladimirteplogaz, Murom Branch, 110-a, Pervomayskaya str., Murom, 602263, Russia
* Corresponding author: firstname.lastname@example.org
The results of the development and experimental verification of a system for obtaining and processing diversity acoustic noise control data are presented. The system is implemented using the LabView G graphical programming language. The length of the measured realization is determined by the ratio of the sample size to the sampling frequency. The number of measurements of each sound signal is determined by the ratio of the analysis time to the length of the measured realization. The measured audio signal implementation in the form of 40,000 16-bit binary numbers is processed using the fast Fourier transform. The result is a discrete spectrum of the audio signal in the range from 0 to 20 KHz with a step of 1 Hz. The results of the system check using a laboratory acoustic chamber and a noise shield are presented. The advantages and disadvantages of the system are formulated.
© The Authors, published by EDP Sciences, 2019
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.